Base Unit Power Settings: 500 mW, 250 mW, 100 mW Probes: Gallium-Aluminum Arsenide infrared, 867 nm, and red. 670 nm, with a minimum output power of 500 mW as measured at the lens.
Probe spot size: 1.04 cm2.
Probe dimensions: 1.25″ Diameter x 7″ long.
Power Meter: Accuracy within 2%. Allows one to check the probe’s power output before beginning treatment.
Dosimetry: An audible tone signifies each Joule of energy output, allowing easy and accurate measurement of the dosage delivered.
Pulsing Mode: Users can choose from preset, extremely low frequencies – 2, 5, 8, 10, 20 Hz and Nogier frequencies – 73, 146, 292, 584, 1168, 2336, 4672, 9344 Hz.
Pulse Width: When operated in pulsed mode, the beam is interrupted for 2 µsec. At the continuous wave setting, there is no pulse interruption.
Laser Class: BRH Class IIIb.
Power Supply: Wall mounted unit. Input: 120VAC @60Hz, 1.3A; Ouput:12VDC, 0.325A.
Lithium Ion Batteries: Are used to power the portable version, permitting the therapist to move from place to place without having an electrical outlet available. The batteries last approximately 10 – 12 hours of use prior to needing recharge. The recharge takes place in approximately 8 hours.
Two pair of safety goggles are provided: Blocking 99% of laser light between 904 and 670nm.
Warranty: 1 year.
FDA Clearance: Class II device.